The 515 FP03 application caters for custody transfer measurement of petroleum products. The frequency flow inputs can accept a quadrature signal for ISO 6551 level B pulse security. An analog temperature and/or density input allows for volume correction to reference conditions..
This instrument is compatible with a wide range of flowmeter frequency outputs, including millivolt signals, reed switches, Namur proximity switches and pulse trains via its smart front-panel program selection.
The flow computer displays the net and gross flowrates, resettable and accumulated totals. It can be used to measure a range of crude and refined petroleum fluids included gasolene, jet fuels, heating oils, diesels, lube oils and LPGs.
The volume total and flowrate are derived from accurately measured frequency and the number of received pulses.
volume = pulses/k-factor
volume flow = frequency/k-factor
The volume correction calculations are based on the ASTM D1250-04 and API Table 54 standard for the following products:
Crude Oils Lube Oils Refined Products Light Hydrocarbon Liquids (LPG) Volume correction for other fluids can be calculated by the following means:
General Coefficient of Expansion Reprogrammed User Table